Vacuum pumps are used in a wide variety of experimental set ups. If vacuum pumps are not properly installed, trapped and exhausted they may expose you to hazardous chemicals and vapors.
- Belt guards must be in place.
- Service cords and plugs must be free of defects.
- Plug the pump directly into an outlet. Do not connect to an extension cord or power strip. See the Electrical Safety Fact Sheet for more information.
- Locate in a vacuum pump cabinet (if one is available) or other ventilated cabinet.
- Place the pump on a tray so that spilled oil is contained.
- Shield any glassware under vacuum.
- Pump oil may be contaminated and should be disposed as chemical waste.
- Pump oil must be compatible (i.e. do not use hydrocarbon pump oil with oxidizing gases or vapors) with the vapors that will pass through the pump.
- Use a cold trap to prevent the degradation and contamination of pump oil.
A cold trap must be placed between the pump and the experiment to minimize the amount of volatile chemicals from reaching the pump oil. Pump oil will break down if exposed to high concentrations of solvents from the vacuum line. This can result in pump damage.
The trap must be adequately sized and cold enough to condense vapors in the experiment. Extreme care must be exercised to prevent the introduction of room air into a trap containing liquid nitrogen. Liquid Nitrogen condenses oxygen and this may cause an explosion. See SOP: Cryogens and Dry Ice for more information about preventing the formation of liquid oxygen and what steps to take if you observe that oxygen has condensed.
Maintain trap during the experiment. Check frequently for blockage.
Use with Hazardous Chemicals
Vacuum pumps which are used to evacuate systems containing toxic, corrosive or volatile substances must be vented into the building’s exhaust system. Failure to properly vent the pump can result contamination of the lab with hazardous chemical vapors.
Use an oil mist separator (purchased from the pump vendor) to prevent oil loss.
Connect the exhaust line to the exhaust port in the vacuum cabinet (if one is present) using a thimble connection or run the hose into the fume hood.
Note that some oil mist separators may not fit into your vacuum pump cabinet. Measure before you order.